With the top off I could check the meter. There are two adjustments, one for slope and the other for overall sensitivity. The slope adjustment (millivolts change per change in exposure EV) is set by a trimmer pot. The manual specifies a setting of exactly 226mV at the indicated test points. This one came out to 225.9mV so no adjustment was needed. The overall sensitivity can be set once the top is back on.