It sounds like your meter is working properly. That's a good reliable CT meter.
Drop in CT for flash units is usually caused by material being deposited inside the tube as it's used.
The coating usually has a slight reddish cast, which lowers the CT of the flash.
I used to be a warranty service station for Novatron, and had occasion to replace flash tubes for that reason.