First check each of the thermometers for any column separation. Then if they are OK place them in an ice-water bath. Each thermometer must be inserted in the bath up to their marked immersion depth. Good lab thermometers should be accurate to plus/minus 1oC. To avoid any parallax error your eye should be immediately above the 0o mark.