I don't have the step wedge shown in the manual. The step wedge I have is the one I used with my old MacBeth and as far as I know, it was made by the previous owner. It is more of a test strip really, with 9 steps from 0.04 to 1.77. Not really enough I don't think. I have a Stouffer step wedge which I read with my MacBeth. I still have those values (21 steps, 0.05 to 3.05).
Originally Posted by ic-racer
I am still baffled by the calibration procedure. I only want to use this device for b&w.
Does anyone in APUG own one of these? I think I need to be walked through this procedure. It sure isn't intuitive.
Last edited by Doc W; 10-10-2013 at 11:09 AM. Click to view previous post history.